Porter A-2689-000 O2 Outlet Primary Check Valve Slotted Back
Brass and chrome plate construction
Accurate diaphragm design
Replacement for slotted Porter primary check valves
You must determine if you have non-slotted or slotted back check valves to order the proper part. The surest method to do this is to remove one from your system to visually check. Sometimes they were installed with faceplates giving improper series numbers. To do this you must first depressurize your system (if you are unsure on how to do this we suggest you enlist the services of a certified medical gas professional). Remove the faceplate, then turn out the check valve turning clockwise.
This is a left-handed thread. To install the check valve reverse the procedure. Proper depth of the check valve will very slightly pull the faceplate tight to the wall. Check to make sure your duplex quick connect operates correctly. If it doesn’t you will need to adjust the depth of the check valve so that the quick connect engages the lip on the check valve.
If an item appears on backorder, place your order and we’ll notify you in a timely manner when to expect delivery. And we’ll ship your item directly from the manufacturer.
Working closely with customers for over 40 years, Broward A&C has always risen to meet the needs of our customers. Our knowledge of all products enables us to provide outstanding customer service, always.